- Product Details
- Company Profile
Scope of application
Saturated vapor pressure tester applicable to GB/T8017-87. Determination of the vapor pressure of petroleum products (Reid method) The Saturated vapor pressure tester volatile crude oil and other volatile petroleum products. But not for the determination of the vapor pressure of the LPG. The sample is measured at
1.Saturated vapor pressure of petroleum products includes the air chamber, single opening petrol chamber The petrol chamber, water bath temperature control device, agitator, pressure gauges and other two openings.
2.The air chamber with a single opening petrol chamber, dual opening The gas chamber by copper material.
3. The water bath for insulation material filling sandwich cooler of installed The main shop electric heater and electric bath mixing device, The bath temperature remains
4.Temperature control device: temperature control accuracy of 37.8 ±
5. Gauge: to test The Reid vapor pressure under different pressures, The instrument is equipped with various range of standard pressure gauge group.
1. Power supply: AC 220V±10%,50Hz;
2. Heating power of water bath: 1600 W;
3. Operation temperature of water bath: room temperature~
4. Temperature controlling accuracy: ±
5. Pressure meter accuracy: ±0.4%;
6. Ambient temperature: -10~
7. Relative humidity: ≤85%;
8. Total power consumption: not more than 1700 W
please leave your contact details when send inquiry!
Send your message to this member
Standard Gauge, ABS Case
Vortex & Swirl Flowmeter (YHLX-1)
Tire Pressure Gauges - A107
Tire Pressure Gauges - A103
Pressure Gauge (X006)